Water Spray Test Equipment
Salt Mist Test Chamber/Materials :
1.laboratory using PVC PVC board
2.test chamber sealed with impact-resistant acrylic panel , 5mm thick
3.supplementary reagent water bucket with hidden observation window ,easy to clean,easy to break .
4.saturated air barrels SUS # 304 stainless steel high pressure barrel insulation best .
5.Blue frame mining laboratory frame indexing plane , the angle can be adjusted , exactly four corners by fog and mist , the test piece is placed and number.
Corrosion Test Meter/Technical parameters:
1.Model : SH - 160
2.laboratory space (W*D*H) cm: 160*100*50
3.Machine External dimensions (W*H*D) cm: 230*130*140
4.test chamber temperature NSS ACSS 35 celsius degree about 1 celsius degree (CASS 50 celsius degree about 1 celsius degree )
5.saturated air barrel temperature NSS ACSS 47 celsius degree about 1 celsius degree (CASS 3 celsius degree about 1 celsius degree )
6.Power supply : AC380V 3*15A
7.test chamber volume : 800L
8.brine volume : 65L [rachel(at)aslitesting.com]
9.Weight : 160kg
Salt Fog Test Chamber/Features:
1.The automatic or manual water system, low water level can be automatically added feature , the test is not interrupted.
2.Precision glass nozzle (PYREX) mist spread evenly and naturally fall on the specimen , and to ensure that no crystalline salt block.
3.with dual over- temperature protection , water shortage alert to ensure safe use .
4.use digital display temperature controller , PID control, error about 0.1 celsius degree .
5.laboratory using direct steam heating, heating up fast and evenly, reducing standby time.
6.spray tower conical diffuser , a guide fog, adjust the volume and even off the fog volume and other functions.
7.saturated air barrel with Henry's law, in order heated humidification, and provide the necessary laboratory humidity.
8.salt concentration : 5 percent; PH value of water is 6.5 - 7.2 between .
ASLI more new products , please attention :
Temperature and humidity chamber
Thermal shock test machine
Enter type high low temperature test chamber