Salt Fog Test Chamber for bearings
Salt spray test machine.are for the surface treatment of a variety of materials.including quality paint.electroplating.organic and inorganic coating, anodizing.corrosion rust treatment tests were carried out to determine the corrosion of time .
Salt Fog Corrosion Test/Technical parameters:
Test Standard:NSS, ACSS CASS
Test temperature:35 about 1 celsius degree or50 about 1 celsius degree
Internal Dimensions:160*100*50 (W*D*H) cm
Dimensions:230*120*150 (W*D*H) cm
Spray Test Equipment/Features:
1 . PVC plate entire mining system.the internal use of advanced three-dimensional reinforcement ring with technology.strong structure, solid.never deformation.
2 . Laboratory with five heating surface heating method can really grasp the average temperature of the test chamber (about 0.1 celsius degree)
3 . Nozzle: Bo Nute Theorem draw salt water spray, fog will make the diffusion rate is doubled, and naturally fall on the specimen.the difference may be controlled falling mist about 0.3ml/80c m h.
4 . Saturated air with Henry's law, be heated humidification, and provide the necessary laboratory humidity.
5. Temperature directly affects the corrosion rate of the conditions precedent.the Company adopted the approach of PID temperature controllers, control error within about 0.5 celsius degree.
6. The only warning devices using low test solution.the test solution when only enough for 5 hours and then spray the buzzer will notify users supplement to ensure the continuous spray test .
7.taken full detection system.when a fault occurs.the fault indicator lights automatically shut down.
Water Spray Test Equipment/Routine maintenance:
1.before using salt spray test machine.the first row of the fog to consider installing the pipe, preferably with a straight wall exhaust pipe through the wall to avoid the indoor salt spray corrosion damaging other equipment.bending too much, too fog effects will affect the row.there will be pressure to produce laboratory.affect the test results.
2, add salt to be careful.not too fierce.saline into the control panel.can cause corrosion and electrical control system instrumentation shorted.causing an accident.
3.over-temperature alarm, please immediately notify the manufacturer to send technical personnel to repair.without delay. Otherwise, burn the box.or electronic control systems.causing an accident.
4.the laboratory should periodically clean up the drainage of water.not a long-term clean-up.water in the salt content will be more and more concentrated.have a corrosive effect on the heat pipe heat pipe will reduce the life .
5.please adjust the size of the spray volume to achieve spray tower.non-adjustable pressure valve.
6.the machine's transparent panel cover and superstructure unavailable corrosive solvent scrubbing.
7.glass filters and protective glass nozzle ( must not use needles or any hard object to sparsely -pass filter or nozzle ) .
8.the end of each test or test solution standard measuring cup should be drained and cleaned in a timely manner in order to avoid accumulation of crystals of salt solution.the impact of settlement calculations.
9.it is recommended to clean equipment regularly test box ( including:spray chamber.a salt solution chamber.preheat tank and sealed tank ) to keep clean equipment.