|Place of Origin:||China|
|Packaging Details:||standard wooden case|
|Supply Ability:||10 sets per month|
|Minimum Order Quantity:||1 set|
Programmable UV Aging Tester
UV Aging Equipment/Technical parameters:
1.Wavelength range: UV-A wavelength range of 320 ~ 380nm
2.radiation intensity: 0.7W/m2/340nm adjustable within
3.Temperature Resolution: 0.1 degree
4.Light Temperature range: 50 degree ~ 60 degree / temperature tolerance of (+/-) 3 degree
5.blackboard thermometer measuring range: 30 ~ 60 ℃ / tolerance of (+/-) 1 degree
6.irradiation test when humidity range: about 45% ~ 80% R.H.
7.Temperature control: PID self-tuning temperature control mode
8.condensing temperature range: 40 degree ~ 60 degree / temperature tolerance of (+/-) 3 degree.
Uv Testing Machine/Structural features:
1, the inner wall: Imported SUS304 stainless steel
2, Shell: cold-rolled plate electrostatic spraying
3, Seal: silicone seal imports
4, temperature and humidity control mode: BTHC balanced thermostat humidity
5, the sample holder stainless steel base frame with box spring liner composition and done
6, South Korea, intelligent digital controller, PID tuning adjustment, 0.2 precision, LED display, with communication interface
7, heating: stainless steel finned air heating
8, water purification equipment
Uv Accelerated Weather Test Chamber/Safety devices:
1, power overload, passing control back to the overload, short circuit protection
2, ground fault protection, motor overload protection
3, over-temperature protection
4, overvoltage, overload
5, for the protection of equipment, all alarms are automatically cut off the power, and send voice prompts.
Aging Test Chamber With UV Lamp/Use site conditions:
1. Temperature: 15 degree ~ 35 degree
2. Relative humidity: not more than 85% RH
3. Around without strong vibration, no strong electromagnetic fields
4. No high concentrations of dust around and corrosive substances
5. No direct sunlight or other sources of heat radiation
6. No strong airflow around when you need to force the ambient air flow, air flow should not be blowing directly on the box.
7. Chamber should be placed smoothly, maintain the level.