Salt Mist Spray Chamber
Salt Water Spraying Test/Materials :
1.Laboratory using PVC PVC board
2.Laboratory seal with impact-resistant acrylic panel , 5mm thick
3.Supplementary reagent water bucket with hidden observation window , easy to clean , easy to break .
4.saturated air barrels SUS # 304 stainless steel high pressure barrel insulation best .
5.Laboratory blue frame mining plane indexing frame, the angle can be adjusted , exactly four corners by fog and mist , the test piece is placed and number.
Salt Mist Test Chamber/Features:
1.using automatic or manual water system, low water level can be automatically added feature , the test is not interrupted. [sales(at)aslitesting.com]
2.precision glass nozzle (PYREX) mist spread evenly and naturally fall on the specimen , and to ensure that no crystalline salt block.
3.with dual over- temperature protection , water shortage alert to ensure safe use .
4.digital display temperature controller , PID control, error ± 0.1 ℃.
5.laboratory using direct steam heating method , heating up fast and evenly , reducing standby time.
6.spray tower conical diffuser , a guide fog, adjust the volume and even off the fog volume and other functions.
7.saturated air barrel with Henry's law, in order heated humidification, and provide the necessary laboratory humidity.
8.salt concentration : 5% ; PH value of water is between 6.5-7.2 .
Laboratory space (W*D*H)cm:60*45*40
Machine External dimensions (W*H × D)cm:110*60*118
Test chamber temperature:NSS ACSS35 ℃ ± 1 ℃ (CASS50 ℃ ± 1 ℃)
Saturated air barrel temperature:NSS ACSS47 ℃ ± 1 ℃ (CASS63 ℃ ± 1 ℃)
Power : AC220V1Φ20A
Test chamber volume : 110L
Brine volume : 15L
Weight : 60kg
Salt Testing Equipment/Maintenance Matters
1.during the test , such as more than one month , please replace heating the water inside the tank cleaning .
2.the salt solution used as the test is not used more than one week , please do not use so as not to affect the outcome of the tests.
3.as from the next test interval is longer, please complete the test , the test chamber cleaning and heating emissions within the tank
a.water heating tank during the discharge - Open the drain valve
b.Water tank isolation within the emissions - will be uprooted middle silicone plugs
c.Preheat the water within the tank emissions - will be uprooted internal silicone plugs